This VS-MI-100-5-FL-B is an 808nm diode laser stack, which has 5pcs 100w bars with PEEK side plate and aluminum shell. Electrodes are on the back of the stack. This kinds of stack maybe have 3 real bars with 2 decorate bars or 4 real bars with 1 decorate bars or 5 real bars.
It is used for 808nm diode laser hair removal handpiece repairing and replacement. Exp: Aroma diode (3 real bars & 2 decorate bars)
Oriental-laser focused on diode laser bar assembling for more than 16 years and we have seen various alma soprano diode laser stacks.Also we can use the same Jenoptik bar like Alma used to recondition the power the laser soprano power.
|>Chioce A: Send the old broken handpiece to our facatory. We will do the process of checking , repairing, cleanning and etc. Then send the fresh new handpiece back to you.||>Save Money. Replace the broken diode stack only. Low cost and convenient|
|>Chioce B: if you have basic assambly capacity, you can send the diode laser stack only to our factory. We remove the old bar and mount the new bar to recover the laser power of the stack.||>Long life time. we use diode laser bars from Germany. The newest Ausn mounting technology will be applied when repairing. After repair. The handpiece will be as a new one. life time is much longer than before|
|>Choice C: Measure your stack and comfrim the water pipe hole and screw hole position. Then press the button below to buy a ready-made stack which is match your old hadnpiece.||>All technical support available. We major in diode laser mounting, software, spare parts over 16 years. All the technologies about laser hair removal machine are open to you.|
|Application||laser hair remova Handpiece Repair And Replacement exp: Aroma diode|
|Power per submounts||100W|
|Number of submounts in one stack||5bars|
|Central wavelength at 25℃||808 ± 5nm|
|Typical Operation current||85A|
|Maximum Operating current||100A|
|Typical Threshold current||15A|
|Maximum Operating Voltage||<10V|
|Fast Axis Divergence (FWHM)||39º|
|Slow Axis Divergence (FWHM)||8º|
|Operation Condition||non-condemsing atmosphere, cleaning room|
|Typical pitch per bar||1.8mm|
|Flow rate||1.8L/min +-10%|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|
Dimension is upon 6bars stack.