This diode laser hair removal handpiece model V-syneron is made of 8 pcs diode bar with micro channel cooler. Total optical power is 800W. The diode laser bars in this handpiece is VS-MI-100-8-Syneron A.
Whole handpiece is made in China.
Service mode: Repair the broken diode laser in the handpiece. Such as replace new laser stack, test TEC peltier, cleaning and so on.
|>Chioce A: Send the old broken handpiece to our facatory. We will do the process of checking , repairing, cleanning and etc. Then send the fresh new handpiece back to you.||>Save Money. Replace the broken diode stack only. Low cost and convenient|
|>Chioce B: if you have basic assambly capacity, you can send the diode laser stack only to our factory. We remove the old bar and mount the new bar to recover the laser power of the stack.||>Long life time. we use diode laser bars from Germany. The newest Ausn mounting technology will be applied when repairing. After repair. The handpiece will be as a new one. life time is much longer than before|
|>Choice C: Measure your stack and comfrim the water pipe hole and screw hole position. Then press the button below to buy a ready-made stack which is match your old hadnpiece.||>All technical support available. We major in diode laser mounting, software, spare parts over 16 years. All the technology are open to you.|
|Related laser stack which match the 808nm hair removal handpiece model V-syneron||VS-MI-100-8-Syneron A|
Syneron Laser Diode Handpiece Specifications
|Model No||808nm Diode laser hair removal handpiece model V-syneron|
|Application||Syneron elos motif HR 808nm/810nm diode laser hair removal machine|
|Number of submounts||8 Bars|
|Power per submounts||100W|
|Central wavelength at 25℃||808 ± 2nm|
|Typical Operation current||60A|
|Typical Threshold current||15A|
|Maximum Operating Voltage||<12V|
|Fast Axis Divergence (FWHM)||39º|
|Slow Axis Divergence (FWHM)||8º|
|Operation Condition||non-condemsing atmosphere, cleaning room|
|Expected lifetime||>10000 Hours|
|Typical pitch per bar||1.8mm|
|Flow rate||0.9L/min ±10%|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|